Recalling Firm
American National Red Cross, Greater Ozark-Arkansas Region
Becton, Dickinson and Company, BD Biosciences

Initial Firm Notification of Consignee or Public
Two or more of the following: Email, Fax, Letter, Press Release, Telephone, Visit

Distribution Pattern
Worldwide Distribution: US (nationwide) and countries of: Canada and Mexico.
Product was distributed to OLD in Georgia, where it was distributed nationwide.
Distributed to domestic and foreign conference attendees.
Affected geographies: Bahamas, Belgium, Canada, Chile, China, Colombia, France, French Guiana, Germany, Greece, Guadeloupe, India, Israel, Italy, Japan, Kazakhstan, Lithuania, Mexico, Netherlands, Panama, Paraguay, Poland, Portugal, Romania, Russia, Singapore, South Africa, Spain, Turkey, United States (including Puerto Rico, US Virgin Islands, and Northern Mariana Islands), Uruguay.
Worldwide Distribution - US (nationwide) and to the countries of : ARGENTINA, AUSTRALIA, AUSTRIA, BELGIUM, CANADA, CHINA, DENMARK, ESTONIA, FINLAND, FRANCE, GERMANY, HUNGRY, INDIA, INDONESIA, ISRAEL, ITALY, JAPAN, REPUBLIC OF KOREA, MEXICO, NORWAY, POLAND, SAUDI ARABIA, SINGAPORE, SLOVAKIA, SOUTH AFRICA, SPAIN, SWEDEN, SWITZERLAND, TAIWAN and UNITED KINGDOM.
Worldwide Distribution and US Nationwide
Worldwide Distribution - US including AR, AZ, CA, CT, FL, GA, HI, IA, ID, IL, IN, KY, LA, MA, MD, MN, MO, MT, NC, NE, NJ, NM, NV, NY, OH,PA, SC, TN, TX, VA, WA, WI, WV and District of Columbia. Foreign distribution to United Arab Emirates, Australia, Belgium, Brazil, Canada, Switzerland, Chile, China, Colombia, Czech Republic , Germany, Spain, France, England, Georgia, Indonesia, Ireland, Israel, India, Italy, Jordan, Japan, Korea, Kuwait, Lithuania, Myanmar, Mexico, Malaysia, Netherland, Norway, Peru, Philippines, Poland, Qatar, Reunion, Romania, Serbia, Saudi Arabia, Singapore, Slovenia, Thailand, Turkey, Taiwan, and South Africa.
Domestic: TN, IA Foreign: England VA/DOD: None
Worldwide Distribution -- US, including DC and PR; and, the countries of SAUDI ARABIA, LEBANON, GERMANY, HONG KONG, SINGAPORE, CANADA, and JAPAN.
Worldwide Distribution: US (nationwide) including states of: TX, CA, TN, NY, LA, IL, MA, PA, OH, AZ, ID, CT, MI, and KY; and countries of: Algeria, Argentina, Australia, Bahrain, Belarus, Bolivia, Brazil, Chile, China, Colombia, Czech Republic, Denmark, Dominican Republic, Ecuador, Egypt, El Salvador, Finland, France, Germany, Hong Kong, Hungary, India, Iran, Iraq, Ireland, Israel, Italy, Japan, Kazakhstan, Korea, Kuwait, Libya, Lithuania, Malaysia, Malta, Mexico, Moldova, Morocco, Myanmar, Netherlands, Philippines, Poland, Portugal, Puerto Rico, Qatar, Russian Federation, Saudi Arabia, Singapore, Slovenia, South Africa, Switzerland, Taiwan, Thailand, Turkey, Ukraine, United Arab Emirates, United Kingdom, , Venezuela, and Vietnam.
Distributed to one customer in Lake Charles, LA and one customer in Spartenburg, SC.
Nationwide Distribution including CA, CT , FL , WI.
Worldwide distribution to U.S. nationwide, Germany and England.
Worldwide Distribution - US Nationwide and the countries of Canada, Albania, Austria, Australia, Bosnia Herzeg., Bangladesh, Belgium, Brazil, Switzerland, Chile, China, Colombia, Czech Republic, Germany, Denmark, Denmark, Estonia, Spain, Finland, France, Great Britain, Georgia, Greece, Croatia, Hungary, Ireland, India, Italy, Lithuania, Luxembourg, Latvia, Macedonia, Malaysia, Netherlands, Norway, New Zealand, Poland, Portugal, Paraguay, Romania, Sweden, Singapore, Slovenia, Slovakia, Turkey, Taiwan, South Africa, Unit.Arab Emirates
Worldwide Distribution - US (nationwide) Distribution and to the countries of : Canada, Japan and Europe.
AL, AR, AZ, CA, CO, CT, DE, FL, GA, IA, IL, IN, KS, KY, LA, MD, MI, MN, MO, MS, NC, NH, NJ, NY, OH, OK, OR, PA, TN, TX, UT, VA, WI, WV and District of Columbia
Worldwide Distribution. US nationwide (TX, CA, NC, OH, SD, PA, CT, UT, MD, TX, GA), Argentina, Taiwan, Australia, Japan, China, New Zealand, Chile and Colombia.
Worldwide Distribution - US distribution to one distributor in FL.
US Nationwide and internationally, including Canada.

Product Description
EVIS EXERA II Duodenovideoscope OLYMPUS TJF TypeQ180V. This instrument has been designed to be used with an Olympus video system center, light source, documentation equipment, monitors, EndoTherapy accessories (such as a biopsy forceps), and other ancillary equipment for endoscopy and endoscopic surgery within the duodenum.
CORID (amprolium) 9.6% Oral Solution Coccidiostat packaged in 16 oz plastic bottle, 12 bottles per case. NADA 013-149. OTC. UPC 3 50604 66970 8. Distributed by: Merial Limited, Duluth, GA.
Fujinon/Fujifilm Processor EPX 2500 Operation Manual.
Fujifilm Processor EPX 2500 High Definition Endoscopy; (Sales Brochure). Use to observe, record and capture images.
Trident 10 degrees X3 Insert 36 mm ID Catalog #623-10-36I. Hip implant component.
Marketing brochures for the PowerPICC and PowerPICC SOLO catheters
MicroScan LabPro Information Manager System, Version 1.0 to Version 4.11, Catalog No. 10714149, 10714150. Intended to manage both microbial identification (ID) and antimicrobial agent susceptibility testing (AST).
PHENYLephrine HCl, 50 mg added to 250 mL 0.9% Sodium Chloride Injection USP (200 mcg per mL), packaged in 250 ml bags, Rx only, PharMEDium Services, LLC, 12620 W. Airport Blvd., Sugar Land, TX 77478, NDC# 61553-171-50
PHENYLephrine HCl, 40 mg added to 250 mL 0.9% Sodium Chloride Injection USP (160 mcg per mL), packaged in 250 ml bags, Rx only, PharMEDium Services, LLC, 12620 W. Airport Blvd., Sugar Land, TX 77478, NDC# 61553-169-50
Bupivacaine HCl 0.0625%, 250 mL total volume in a 250 mL LifeCare Bag in Sodium chloride 0.9%, Epidural Use Only, Rx Only, PharMEDium Services, LLC, 6100 Global Drive, Memphis, TN 36141, NDC# 61553-197-50
Morphine Sulfate, 1 mg per mL (250 mg per 250 mL), 250 mL total volume in a 250 mL LifeCare bag in Sodium Chloride 0.9%, IV Use Only, Rx only, PharMEDium Services, LLC, 6100 Global Drive, Memphis, TN 36141, NDC# 61553-181-50
Fentanyl Citrate, 5 mcg per mL (1,250 mcg per 250 mL), 250 mL total volume in a 250 mL LifeCare bag in Sodium Chloride 0.9%, IV Use Only, Rx only, PharMEDium Services, LLC, 6100 Global Drive, Memphis, TN 36141, NDC# 61553-107-50
PHENYLephrine HCl, 20 mg added to 250 mL 0.9% Sodium Chloride Injection USP (80 mcg per mL), packaged in 250 ml bags, Rx only, PharMEDium Services, LLC, 12620 W. Airport Blvd., Sugar Land, TX 77478, NDC# 61553-156-50
PHENYLephrine HCl, 60 mg added to 250 mL 0.9% Sodium Chloride Injection USP (240 mcg per mL), packaged in 250 ml bags, Rx only, PharMEDium Services, LLC, 12620 W. Airport Blvd., Sugar Land, TX 77478, NDC# 61553-236-50
PHENYLephrine HCl, 10 mg added to 250 mL 0.9% Sodium Chloride Injection USP (40 mcg per mL), packaged in 250 ml bags, Rx only, PharMEDium Services, LLC, 12620 W. Airport Blvd., Sugar Land, TX 77478, NDC# 61553-149-50
ARTISTE, ONCOR Avantgarde, ONCOR Impression Plus, ONCOR Expression, PRIMUS, PRIMUS Plus, Syngo RT Therapist; therapeutic treatment of cancer Product Usage: The intended use of the SIEMENS branded ARTISTE and ONCOR family of linear accelerator systems is to deliver X-Ray photon and electron radiation for the therapeutic treatment of cancer. PRIMUSTM family of linear accelerator systems is to deliver x-ray radiation for therapeutic treatment of cancer. Syngo RT Therapist is a software application whose indication for use includes the viewing, processing, filming, and archiving of medical images. It also permits patient data management, patient selection/setup, patient positioning verification, treatment planning, treatment delivery/verification, and treatment recording.
Label reads in part as: ''GRAVIOLA EXTRACT***1 FL Oz (30 ml)*** ORIGINAL FORMULA***Manufactured by: Majopa Industries Corp.*** PRODUCT OF PUERTO RICO,USA''. Drug claims in the product label are as follows: ''Researchers have found that Extracts from Graviola leaves can produce acetogenins which are extremely effective in the treatment of cancer cells. Graviola extract can works like an antibiotic against cancer cells by stopping growth of the enzymes needed for cell growth. It can be:cellular protector, anti tumorous, antiviral. Also can be used as: Digestive stimulant, Strengthens immune system, antifungal, tranquilizer''.
The Leica ASP6025 Tissue Processor. Pathology and Hematology Departments. The Leica ASP6025 Tissue Processor is an automated system used to process tissue specimens for examination through fixation, dehydration and infiltration.
3.5mm Non Locking Cortical Screw, in the PediLoc Tibial Plate System Part number 00-0903-2510, 3.5mm x 10mm Non Locking Cortical Screw. Used in pediatric orthopedics.
ORS Fluid Warming and Slush Drapes, Model #'s ORS-100, ORS-110, ORS-130, ORS-300, ORS-301, ORS-320, ORS-321, ORS-325, ORS-330, ORS-331 and ORS-2200. The ORS Fluid Warming and Slush Drapes are individually wrapped and placed in poly-Tyvek pouches with a label insert, then packaged as 24 per case in a labeled corrugated shipper box which is subsequently sterilized by gamma irradiation (ORS-2200 only) or ethylene oxide. Intended for use during various surgeries where warm irrigation, slush and/or cold solution are required.
Dynamic Locking Screw System (DLS) 3.7mm & Dynamic Locking Screw (DLS) 5.0mm; intended for use in long bone fractures, the fixation of osteopenic bone, the fixation of osteotomies, and for the fixation of non-unions and malunions.
Disposable Reflective Marker Spheres The Disposable Reflective Marker Spheres used in conjunction with a stereotaxic instrument consisting of a rigid frame with a calibrated guide mechanism for precisely positioning probes or other devices within a patient's brain, spinal cord, or other part of the nervous system. They are attached to reference arrays and instruments, thus enabling the infrared tracking systems to detect the position of the patient and instruments in the surgical field. Radiology.
Regard (tm), Item Number: 800628; Sterile, CV0805 - CABG B. Used in cardiovascular surgical procedures.
Regard (tm), Item Number: 800208002; Sterile, CV0361B - CABG A&B. Used in cardiovascular surgical procedures.
Terumo Custom Cardiovascular Procedure Kit Tubing Pack P/N: 66041-1 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed.
Terumo Custom Cardiovascular Procedure Kit X-Coated Adult Pak P/N: 73124 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed
Terumo Custom Cardiovascular Procedure Kit X-Coated Perfusion Pack with FX25RE P/N: 7928-03 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed
Terumo Custom Cardiovascular Procedure Kit X-Coated Pediatric Pack P/N: 74039-02 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed
Terumo Custom Cardiovascular Procedure Kit X-Coated Perfusion Pack P/N: 74077-01 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed
Terumo Custom Cardiovascular Procedure Kit X-Coated PED/Adult Main Pack P/N: 75185 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed
Terumo Custom Cardiovascular Procedure Kit X-Coated PED/Adult Main Pack P/N: 75529 Terumo Cardiovascular Procedure Kits are manufactured and customized to the design and specifications of the purchaser and/or the purchaser's principal technical representative for the application intended by the end user. The kits are comprised of a variety of legally marketed medical devices [including Medivators HPH700 High Performance Hemoconcentrators]. The Hemocor HPH700 Hemoconcentrator is intended for use as an ultrafiltration system to remove excess fluid during and/or following cardiopulmonary bypass procedures where acute hemodilution has been employed
Refresh Lacri-Lube (Mineral Oil 42.5% and White Petrolatum 56.8%) Lubricant Eye Ointment, packaged in a) Net wt. 0.12 oz (3.5 g) tubes (NDC 0023-0312-04) UPC 3 0023-0312-04 2 and b) Net wt. 0.25 oz (7 g) tubes (NDC 0023-0312-07) UPC 3 0023-0312-07 3, Allergan, Inc., 2525 Dupont Drive, Irvine, CA 92612.
Refresh P.M. (Mineral Oil 42.5% and White Petrolatum 57.3%) Lubricant Eye Ointment, Net wt. 0.12 oz (3.5 g) tube, Allergan, Inc., 2525 Dupont Drive, Irvine, CA 92612, NDC 0023-0240-04, UPC 3 00230 66704 3.
FML (fluorometholone ophthalmic ointment) 0.1%, 3.5 g tube, Rx only, Allergan, Inc., Irvine, CA 92612, NDC 0023-0316-04, UPC 3 0023-0316-04 0.
BLEPHAMIDE (sulfacetamide sodium and prednisolone acetate ophthalmic ointment, USP) 10 %/0.2 %, 3.5 g tube, Rx only, ALLERGAN, Irvine, California 92612, NDC 0023-0313-04, UPC 3 00230 31304 9.
PenBlade Safety Scalpel, for the incision of human skin and tissue, and to cut or trim sutures and catheters up to 6Fr. Models PB-M-10, PB-M-11P, PB-M-15
FRESH ORIGINS Microgreens Celery NET WT 4 OZ San Marcos, CA 92069 packed in clear plastic clam shell.
Siemens RAPIDPoint 500 Measurement Cartridge Lactate (750 tests) Part Number: 10491449 Product Usage: The RAPIDPoint 500 system is designed for professional use in a point-of-care or laboratory environment. This system tests blood gases, electrolytes, metabolites, total hemoglobin, and hemoglobin derivatives in arterial, venous, and capillary whole blood samples. The following parameters are tested: pH, pC02, p02, Na+, K+, Ca++, Cl-, glucose, lactate, tHb, F02Hb, FCHb, FMetHb, FHHb, nBili
Siemens RAPIDPoint 500 Measurement Cartridge Lactate (400) tests Part Number: 10491448 Product Usage: The RAPIDPoint 500 system is designed for professional use in a point-of-care or laboratory environment. This system tests blood gases, electrolytes, metabolites, total hemoglobin, and hemoglobin derivatives in arterial, venous, and capillary whole blood samples. The following parameters are tested: pH, pC02, p02, Na+, K+, Ca++, Cl-, glucose, lactate, tHb, F02Hb, FCHb, FMetHb, FHHb, nBili
Siemens RAPIDPoint 500 Measurement Cartridge Lactate (250) tests Part Number: 10491447 Product Usage: The RAPIDPoint 500 system is designed for professional use in a point-of-care or laboratory environment. This system tests blood gases, electrolytes, metabolites, total hemoglobin, and hemoglobin derivatives in arterial, venous, and capillary whole blood samples. The following parameters are tested: pH, pC02, p02, Na+, K+, Ca++, Cl-, glucose, lactate, tHb, F02Hb, FCHb, FMetHb, FHHb, nBili
Siemens RAPIDPoint 500 Measurement Cartridge Lactate (100) tests Part Number: 10844813 Product Usage: The RAPIDPoint 500 system is designed for professional use in a point-of-care or laboratory environment. This system tests blood gases, electrolytes, metabolites, total hemoglobin, and hemoglobin derivatives in arterial, venous, and capillary whole blood samples. The following parameters are tested: pH, pC02, p02, Na+, K+, Ca++, Cl-, glucose, lactate, tHb, F02Hb, FCHb, FMetHb, FHHb, nBili
ThruPort Knot Pusher. This device is sold individually (Model KP1) and as part of the ThruPort Valve Placement Pack (Model VPP). packaged sterile and non-pyrogenic in a sealed, peel-type pouch. The Knot Pusher is a sterile disposable plastic rod used to advance extracorporeal suture knots during minimally invasive cardiac surgery.
Sam Mills Gluten Free Chocolate Chip Granola Bars with Dark Chocolate 5 bars/4.4oz Box (0.88oz each bar)
Smoked Tofu is tofu sandwich on bread or a bun, enclosed in a clam shell with label
New York Deli Tofu is is tofu sandwich on bread or a bun, enclosed in a clam shell with label
Jamaica Spice Tofu is Tofu sandwich on bread or a bun, enclosed in a clam shell with label
Jamaican Spice Tempeh is tofu sandwich on bread or a bun, enclosed in a clam shell with label
Seitan Steak is tofu sandwich on bread or a bun, enclosed in a clam shell with label
Tofustrami is tofu sandwich on bread or a bun, enclosed in a clam shell with label
0.9% Sodium Chloride Injection, USP, 100 mL MINI-BAG VIAFLEX Container, Single Dose Container, Rx Only, Baxter Healthcare Corporation, Deerfield, IL 60015, USA, NDC 0338-0049-38
CD4 (SK3), FITC; Contains CD4 FITC with gelatin and 0.1% sodium Azide. Catalog number 340133 in vitro diagnostic used in Hematology: The SK3 monoclonal antibody reacts with human CD4, a 59-kDa cell surface receptor expressed by a majority of thymocytes, a subpopulation of mature T helper cells, and at low levels on monocytes. CD4 is a receptor for the human immunodeficiency virus (HIV).
1% Lidocaine HCl Injection, USP, 10 mg/mL, 30 mL Vial, Single-dose Preservative Free, Rx only, Hospira, Inc., Lake Forest, IL 60045, USA, NDC 0409-4279-02
Contra-Angles CA1:1L EVO 15 Catalog numbers: 1600939-001 and 1600938-001 Used for generally dentistry procedures
Synthes Cranial Flap Tube Clamp and Crimping Device Product Usage: The Synthes Cranial Flap Tube Clamp System is intended to reattach a cranial bone flap to the surrounding cranium after a craniotomy procedure.
Plasma Frozen within 24 Hours After Phlebotomy
Red Blood Cells Leukocytes Reduced Irradiated

Code Info
The brochures were purchased from the vendor on 04/14/2015, Code S120667R0
Part number 00-0903-2510, Lot number 150192-J
Model numbers: 41773G Disposable Reflective Marker Sphere (90 Pcs.) 41774G Disposable Reflective Marker Sphere (270 Pcs.) 41772G Disposable Reflective Marker Sphere (3 Pcs) Catalogue numbers: 41773G Disposable Reflective Marker Sphere (90 Pcs.) 41774G Disposable Reflective Marker Sphere (270 Pcs.) (Note: 41772G is not available as single item for sale, but contained in the above 41773G or 41774G) Lot Numbers: 1014213001, C103461502, C107761402, C100111501, C1034813001, C108341301, C101021502, C103881402, C108721402, C101161401, C104761402, C109051301, C101611502, C105541402, C109251402, C102311401, C106521402, C1098312001, C102481502, C106721301 and C110411401.
Serial Numbers: 2517102517 through 2519000012. Cartridges within this serial number range that have a green dot on the box and/or the cartridge are not affected
08 30 (Use By August 30th) 09 02 (Use By Sept. 2nd)
Catalog Number 340133; lot # 5112711, Expiration date: 30 Nov 2016.
Catalog numbers: 1600939-001 and 1600938-001 Serial numbers concerned: - Ref: 1600939-001: Serial numbers included in the following range: 15B0001 up to 15F0011 - Ref: 1600938-001: Serial numbers included in the following range: 15B0007 up to 15F0027

Product Quantity
U.S. Quantity Distributed: 163 systems; Foreign Quantity Diistributed: 410 systems.
For 28 US consignees, 9,900 Disposable reflective marker spheres ***For 451 non-US consignees, 359,280 Disposable reflective marker spheres

Reason for Recall
New and validated reprocessing procedures have been issued for the Olympus TJF-Q180V duodenoscope (''TJF-Q180V''), consisting of revised manual cleaning and high level disinfection procedures.
Product container does not conform to approved application.
Fujifilm Medical Systems, USA, Inc. (FMSU-ESD) has become aware that the EPX-2500 Operation Manual and sales brochure incorrectly identify that the ED-530XT endoscope is meant to be used with the EPX-2500 Processor. The Operation Manual and sales brochure state that when the ED-530XT endoscope is used with the 2500 processor, a ''super image'' is displayed when it is actually a ''standard image'' that is displayed.
it was discovered that item 623-10-36I Lot 3Y0KHP, although meeting all device specifications, was determined to be unacceptable due to a documentation issue. However the parts were inadvertently shipped to the field.
Bard Access Systems is recalling marketing brochures for the PowerPICC and PowerPICC SOLO catheters due to incorrect indication for use contained in the brochure.
Beckman Coulter is recalling the MicroScan LabPro Information Manager System because the software incorrectly allows the operator to manually edit the carbohydrate substrates when manually reading dried overnight gram negative panels with an ID Hold status.
Potential for expulsion of the Power Supply on Lightspeed CT scanner or Discovery ST PET CT or Discovery LS PET CT scanner.
Presence of Particulate Matter: The firm manufactured products using Hospira 0.9% Sodium Chloride, USP injection which were subsequently recalled due to the presence of particulate matter (human hair).
A software fix has been released to prevent automatic movement resulting in a collision safety risk for patients.
Firm's dietary supplements were being promoted for conditions that caused the products to be drugs under the FD&C Act.
The ASP6025 Tissue Processor instrument is incorrectly getting into a state of bottle empty during retort filling, after passing the fill lever sensor 2 and prior to reaching level sensor 3, thus not completing a 5 litre fill.
Misbranded Red Blood Cells were distributed.
The 00-0903-2510, 3.5mm x 10mm Non Locking Cortical Screw, Lot # 150192-J package may contain a different screw. The screw in the package is a 00-0903-2526, 3.5mm x 26mm, and is etched correctly.
This recall is being conducted because the sterile packaging may contain small (less than 250 microns wide) channels in the seal of the pouch. If channels are present, the package integrity is compromised and there exists a remote chance that a non-sterile drape may be used during a surgical procedure.
DLS pin breakage during planned implant removal, after uneventful and successful healing of the fracture. This notice to additional consignees identified after recall Z-0390-2014, related to complaints of breakage located at the bottom of the screws and under the locking screw head.
Disposable Reflective Marker Spheres (DRMS) for Brainlab Image Guided Surgery (IGS) Systems: DRMS spheres may separate at the mid-point where the two halves of the sphere are sealed together.
Surgical kits contain a surgical scrub product which may be chemically contaminated and lack sterility assurance.
Custom Cardiovascular Procedure Kits contain recalled component Medivators HPH700 High Performance Hemoconcentrators
Presence of Particulate Matter: Due to customer complaints of small black particles, identified as part of the ointment tube cap, generated by the action of unscrewing the cap from the aluminum tube and potentially introducing the particle into the product.
Cracks were discovered in the PET tray of a small population of sterile product. These cracks create a risk of loss of sterile barrier for the product. Sampling and root cause analysis indicated that the risk could extend throughout the lots manufactured from September 2014-July 2015. This led to the decision to recall the product.
RAPIDPoint 500 measurement cartridge serial numbers 2517102517 through 2519000012, negative sodium bias on both QC and patient results
Edwards Lifesciences is recalling certain lots of ThruPort Systems Knot Pusher (Model KP1) and ThruPort Systems Valve Placement Pack (Model VPP) because the configuration of the slot at the tip of the knot pusher may inhibit intended ease of use. Fraying or splitting of the suture could occur, which could compromise the integrity of the suture knot.
Product contains wheat bun but the ingredients statement does NOT declare wheat and the Contains statement declared Soy only.
Wheat bun is declared in the Ingredients statement but the Contains statement declared Soy only.
Lack of Assurance of Sterility: Recalling firm reported a complaint for mold on the interior surface of the overpouch.
CD4 FITC label contains an error in the Spanish small text - it states CD8 FITC.
Presence of Particulate Matter: particulate matter identified as iron oxide, was found embedded in the neck of glass vials.
Bien-Air Dental SA is recalling Contra-Angles CA1:1L EVO 15 because of possible loosening of the dental tool, which may lead to potential swallowing of it.
The Synthes Cranial Flap Tube Clamp and Crimping Device for Cranial Tube Clamp had been labeled MR Safe although they do not have the testing protocols currently required to designate them as MR Safe.
Blood products, collected from an ineligible donor, were distributed.
Blood Products, lacking assurance of collection in an aseptic manner, were distributed.